Then try after launching the software. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab. The computer put the serial adapter on com 8 – which would not do. If serial, the USB adapters often do not work. Helpful 1 Not Helpful Comment Flag. Please enter a valid email address.
|Date Added:||25 May 2016|
|File Size:||32.49 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
That should fix it. Seller assumes all responsibility for this listing.
If serial, the USB adapters often do not work. You should hiproo your serial adapter in there. This is odd, the hi-pro should be on the serial adpater’s com port.
GN Otometrics Warns of Imitation HI-PRO Devices on Market
Please enter a valid email address. Then try after launching the software. An older version of ActiveSync should not make a difference. Instant Repair Seriap over the Phone 6ya.
Hipro MADSEN Electronics GN Otometrics Hi-pro Serial/usb Interface | eBay
The max point reward for answering a question is The procedure for selecting the com port is the same on earlier versions of windows. Invalid serial numbers, irregular CD labeling, sub-standard bumpers on the bottom of the housing, and general inferior workmanship are just a few of the signs that can identify the fake HI-PRO products by simple visual inspection, says Otometrics.
According to the company, the illegal copy products appear similar to the original HI-PRO products but are of poor quality, do not have the same functionality as the original HI-PRO products, and are not covered by any warranty or service. Back to home page Return to top. I just set up a serial Hi-Pro box for one of my customers on their new laptop.
Hipro MADSEN Electronics GN Otometrics Hi-pro Serial/usb Interface
Also, try installing ActiveSync on another computer that has a serial port and try connecting the Jornada to that computer. Also the software i was using – Noah, with a beltone module for programming hearing aids. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
The facilities are the same, just slightly rearranged. Your answer needs to include more details to help people. In addition, they are not compatible with authorized support and service tools, and could potentially invalidate fitting procedures. Hearing care professionals who suspect they may have a copy product are encouraged to contact serjal seller and return the device.
For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees. Uniden G Guardian Wireless It may be a problem with your computer. I think this puppy is toast. If your computer has an infrared port you could try that, I have had a lot of success with infrared connections with my Jornada. The support page for Keyspan USB devices is here: The drivers available so far are listed here: Give your patients the open….
Skip to main content. How many hearing aids are now rechargeable, and what can a hearing aid user do to ensure the rechargeable battery lasts all day? Posted on Aug 05, How long should the hearing aid battery last after a full charge, and how does Bluetooth affect this?